EA 9777-2 Page 2 ELECTRONIC ASSEMBLY reserves the right to change specifications without prior notice. Printing and typographical errors reserved. Documentation of ...
Report CopyRight/DMCA Form For : Operation Manual Ee310 Epluse Com
E E Elektronik Ges m b H provides no warranty of any kind for this publication and accepts no liability for. improper use of the products described, This publication might contain technical inaccuracies or typographical errors The content is revised regularly and is. not subject to change management The manufacturer reserves the right to modify or change its described products. at any time,Copyright E E Elektronik Ges m b H,All rights reserved. FCC notice, This equipment has been tested and found to comply with the limits for a Class A digital device pursuant to part 15. of the FCC Rules These limits are designed to provide reasonable protection against harmful interference when. the equipment is operated in a commercial environment This equipment generates uses and can radiate radio. frequency energy and if not installed and used in accordance with the instruction manual may cause harmful. interference to radio communications Operation of this equipment in a residential area is likely to cause harmful. interference in which case the user will be required to correct the interference at his own expense. ICES 003 Issue 5,CAN ICES 3 A NMB 3 A,1 General 4,1 1 Explanation of symbols 4. 1 2 Safety instructions 4,1 2 1 General safety instructions 4. 1 2 2 Alarm module with voltages 50 V option AM2 4. 1 2 3 Integrated power supply 100 240 V AC option AM3 4. 1 2 4 Mounting start up and operation 4,1 2 5 Intended Use 5. 1 2 6 Disclaimer 5,1 3 Environmental aspects 5,1 4 ESD Protection 5. 2 Scope of supply 6,3 Product design 6,3 1 Dimensions 7. 3 1 1 Enclosure 7,3 1 2 Probes 7,4 Installation 8,4 1 Mechanical installation 8. 4 1 1 Mounting of the polycarbonate enclosure 8, 4 1 2 DIN rail mounting of the polycarbonate enclosure 8. 4 2 Electrical connection 9,4 2 1 Electrical connection and wiring 10. 4 3 Probe mounting wall duct version 13,4 4 Mounting remote sensing probe of EE310 T5 13. 4 5 Mounting pressure tight probe of EE310 T10 14, 4 5 1 Installation of the probe directly in the process see Fig 10 14. 4 5 2 Installation of the probe with ball valve set 14. 5 Optional modules 16,5 1 Alarm module option AM2 16. 5 2 Integrated power supply 100 240 V AC option AM3 17. 5 3 RS485 Module Modbus RTU option J3 18,5 4 Ethernet Module Modbus TCP option J4 18. 5 4 1 Available TCP and UDP ports 18,5 4 2 IPv4 Settings 19. 5 4 3 Retrofit with Ethernet Module 20,5 5 Modbus Map 20. 5 6 Pluggable probe option PC4 21,6 Operation 22,6 1 Configuration interface 22. 6 2 3 5 TFT Colour Display 23,6 3 Chart Data logger 23. 6 4 Configuration Menu 24,6 5 Status information 25. 6 6 Buzzer ON OFF 25,6 7 Error indication 25,7 Maintenance 26. 7 1 Self diagnosis and error messages 26,7 1 1 Error messages via display 26. 7 1 2 Error messages via LEDs 26,7 1 3 Solving typical problems 27. 7 2 Cleaning the sensing head filter replacement 27. 7 3 RH T adjustment and calibration 27,8 Replacement parts Accessories 28. 9 Technical data 29,10 Appendix 30,10 1 Overview 30. 10 2 Detailed information 30,10 3 Optional menu 33. Operating instructions for EE310 Humidity Temperature Transmitter 3. This operation manual serves for ensuring proper handling and optimal functioning of the device. The operation manual shall be read before commissioning the equipment and it shall be provided to all. staff involved in transport installation operation maintenance and repair. The operation manual may not be used for the purposes of competition without the written consent of. E E Elektronik and may not be forwarded to third parties Copies may be made for internal purposes. All information technical data and diagrams included in these instructions are based on the information. available at the time of writing,1 1 Explanation of symbols. This symbol indicates safety information, It is essential that all safety information is strictly observed Failure to comply with this. information can lead to personal injuries or damage to property E E Elektronik. assumes no liability if this happens,This symbol indicates instructions. The instructions shall be observed in order to reach optimal performance of the device. 1 2 Safety instructions,1 2 1 General safety instructions. Avoid any unnecessary mechanical stress and inappropriate use. When replacing the filter cap make sure not to touch the sensing elements. For sensor cleaning and filter cap replacement please see Cleaning instructions at www epluse com. Installation electrical connection maintenance and commissioning shall be performed by qualified. personnel only, The devices are designed for the operation with class III supply EU and class 2 supply NA. The power supply must be switched off before opening the housing. An existing Ethernet connection must be disconnected before opening the housing. 1 2 2 Alarm module with voltages 50 V option AM2, The optional alarm module is isolated from the low voltage side of EE310 by a special partition this. must remain fitted at all times in the back section of the enclosure. The EE310 enclosure must be tightly closed during operation An open enclosure corresponds to IP00. and exposes components carrying dangerous voltage Any work maintenance for instance on the. device may be performed by qualified staff only, 1 2 3 Integrated power supply 100 240 V AC option AM3. The EE310 enclosure must be tightly closed during operation An open enclosure corresponds to IP00. and exposes components carrying dangerous voltage Any work maintenance for instance on the. device may be performed by qualified staff only,1 2 4 Mounting start up and operation. The device has been produced under state of the art manufacturing conditions has been thoroughly. tested and has left the factory fulfilling all safety criteria. The manufacturer has taken all precautions to ensure safe operation of the device The user must en. sure that the device is set up and installed in a manner that does not have a negative effect on its safe. The user is responsible for observing all applicable safety guidelines local and international with re. spect to safe installation and operation on the device This operating manual contains information and. 4 Operating instructions for EE310 Humidity Temperature Transmitter. warnings that must be observed by the user in order to ensure safe operation. Mounting start up operation and maintenance of the device may be performed by qualified staff only. Such staff must be authorized by the plant operator to carry out the mentioned activities. The qualified staff must have read and understood this operating manual and must follow the instruc. tions contained within, All process and electrical connections shall be thoroughly checked by authorized staff before putting. the system into operation, Do not install or start up a device supposed to be faulty Make sure that such devices are not acciden. tally used by marking them clearly as faulty, A faulty device may only be investigated and possibly repaired by qualified trained and authorized. staff If the fault cannot be fixed the device shall be removed from the system. Service operations other than described in this operating manual may only be performed by the. manufacturer,1 2 5 Intended Use, EE310 is optimized for reliable measurement in demanding industrial applications In addition to highly. accurate measurement of relative humidity RH and temperature T the transmitter also calculates. parameters such as dew point absolute humidity and mixing ratio. Various models are available including wall duct and remote probe The remote probe can be used up. to 180 C 356 F and the pressure tight probe up to 20 bar 290 psi The use of the EE310 other than. described in this manual is not allowed, The manufacturer cannot be held responsible for damages as a result of incorrect handling installation. and maintenance of the device, Unauthorized modifications of the product lead to loss of all warranty claims The device may only be. powered as decried in this manual,1 2 6 Disclaimer. The manufacturer or his authorized agent can only be held liable in case of willful or gross negligence. In any case the scope of liability is limited to the corresponding amount of the order issued to the ma. nufacturer The manufacturer assumes no liability for damages incurred due to failure to comply with the. applicable regulations operating instructions or the operating conditions Consequential damages are. excluded from the liability,1 3 Environmental aspects. Products from E E Elektronik are developed and manufactured observing of all relevant requirements. with respect to environment protection Please observe local regulations for the device disposal. For disposal the individual components of the device must be separated according to local recycling. regulations The electronics shall be disposed of correctly as electronics waste. 1 4 ESD Protection, The sensing elements and the electronics board are ESD electrostatic discharge sensitive compo. nents of the device and must be handled as such The failure to do so may damage the device by. electrostatic discharges when touching exposed sensitive components. Operating instructions for EE310 Humidity Temperature Transmitter 5. 2 Scope of supply,Included in the scope of supply of. EE310 according to ordering guide all versions,Operation manual all versions. Inspection certificate according to DIN EN 10204 3 1 all versions. Mating plug for integrated power supply AM3,Mating plug RKC 5 7 AM3 E4 E6 E12. Mating plug RSC 5 7 2 pcs for option E12 E5 E6 E12. Mating plug HPP V4 RJ45 Cat5 J4,3 Product design,EE310 T5 T10 EE310 T1. Fig 1 Product design, 1 Front section with electronic display and probe cable. 2 3 5 TFT colour display,3 5 push buttons for configuration menu. 4 Back section with electrical connection alarm supply module and mounting holes. 5 Micro USB service interface,6 Standard cable glands connectors. 7 Additional cable gland connector,8 Probe cable gland connector optional. Connection terminal block,add on chip,Back section. optional modules,Front section Mounting plate,Fig 2 Modular enclosure. 6 Operating instructions for EE310 Humidity Temperature Transmitter. 3 1 Dimensions,3 1 1 Enclosure,Polycarbonate Stainless steel. 180 7 09 218 8 58,166 6 54 200 7 87,153 6 02 71 186 7 32 73. 3 1 2 Probes,T1 Wall mount T5 Remote probe up to 180 C 356 F. code cable length 1,L2 code probe length 1 15 0 60. ressure tight probe up to 20 bar 300 psi,code cable length 1. T2 Duct mount L2 200 7 87 15 0 60,200 7 87 L2,1 2 ISO or. adjustable 1 2 NPT,min 23 1 max 164 6 5,1 Refer to ordering guide. 2 L filter length refer to data sheet Accessories, Operating instructions for EE310 Humidity Temperature Transmitter 7. 4 Installation,4 1 Mechanical installation,4 1 1 Mounting of the polycarbonate enclosure. Drill the mounting holes according to the corresponding mounting pattern below. Mount the back section of the enclosure with 4 screws screw diameter 4 2 mm 0 2 not. included in the scope of supply,7 0 28 5 1 0 20, Fig 3 Drilling pattern of polycarbonate enclosure in mm inch. 7 0 28 5 1 0 20, Fig 4 Drilling pattern of stainless steel enclosure in mm inch. 4 1 2 DIN rail mounting of the polycarbonate enclosure. Mount the two DIN rail brackets to be ordered separately see chapter 8 onto the back section. Snap in the enclosure onto the DIN rail,Fig 5 DIN rail installation. 8 Operating instructions for EE310 Humidity Temperature Transmitter. 4 2 Electrical connection, The electrical installation of the EE310 shall be performed by qualified personnel only Observe all. applicable national and international requirements for the installation of electrical devices as well as for. power supply according to EN 61140 class III EU and class 2 supply North America. For EE310 with alarm module option AM2 or integrated power supply 100 240 V AC option AM3. the metal housing must be grounded during operation. Connection diagram,12 30 V AC,mA OUT 1 4 Analogue outputs. mA OUT 2 5,GND RS485 6,Modbus RTU,RS485 A D 7,galvanically isolated. RS485 B D 8,Fig 6 Electrical connection,Analogue outputs. Both analogue outputs shall be configured to either voltage or current Measurands analogue output. range and scaling are freely selectable All settings can be performed via display and push buttons or. using the EE PCS Product Configuration Software see EE PCS Product Configuration Software see. chapter 6 1,Error indication, The analogue outputs feature an error indication function according to NAMUR NE43 In the case of an. error the output signal will freeze at 21 mA or 11 V respectively. By default the error indication is disabled, When changing one of the analogue outputs from current to voltage and vice versa the second output. will change automatically as well The output scaling changes automatically if it is out of physical range. i e 20 mA will be changed to 10 V instead of 20 V, Check output scale after changing between voltage and current output. Operating instructions for EE310 Humidity Temperature Transmitter 9. 4 2 1 Electrical,Standardconnection andEE310 EE360. wiring Electrical connection for stainless steel housing HS2. Standard Standard,Option Polycarbonate V 1,Standard enclosure Stainless steel enclosure. Standard Pin assignment,Cable Standard GNDV 21 V 1. glands GNDV 312 V,OUTGND1 23,2x M16x1 5 Standard GND 3. GND RS485 45,6 OUT12 45,Option J4 OUT 1 4,2x M16x1 5 2x M16x1 5. GNDA RS485,RS485 D 2 56,7 GNDOUT 2,Option2xJ4M16x1 5 Option J4. RS485 687 GND RS485 67 2,2x M16x1 5 RS485,D RS485 GND. Option J4 Option,M16x1 5 mounted,power7supply,Option J4 1 xRS485. M16x1 5 enclosed,D 78 RS485,analogue output o,RS485 A D 7. J4 RS485 B D 8 RS485 B D,RS485 B D 8,2x M16x1 5,GND RS485 6. Option E5 Option E6,RS485 A D 7,RS485 B D 8, EE310 EE360 Electrical connection for stainless steel housing HS2. Modbus TCP,Ethernet Ethernet,Standard Option E4,EE310 EE360 Electrical connection. Modbus TCP,Ethernet for stainless,Modbussteel,Ethernet TCP housing HS2. Option E4 TCP,Modbus Modbus TCP,Modbus TCP M16x1 5. Standard Option E4E4,GND Output,Option AM3,power supply. E4 Option E4 Option E4 RTU,Modbus GND Output GND Output. analogue output,Option E4 OUT 2 GND Output,GND Modbus. Modbus TCP3,GND Output,GND OUT 2 GND,OUT 2 45 3 GND OUT 2 4 3 GND. 1 x M16x1 5 enclosed,OUT 1 4 5 3power supply,1 5 2 OUT 1 1 5 2GND. V V Output,1RTU 2 OUT 1,power supply,analogue output V OUT 1 V Modbus. 1 x M16x1 5 mounted,power supply power supply,power supply only with. 1 x M16x1 5analogue,enclosedoutput OUT,power supply. power output,analogue output order2 code J3 GND,power supply. output Option,analogue E5 GND RS485,E5 Option E5,analogue output Option E5. 5GND RS485 power supply,100 240 V AC,GND RS485 shielded. GND RS485OUT 1,Option E5 n c RS485 A V,Option E5 Option E6 shielded. GND RS485 Option AM2,power supply shielded RS485 A. n c n c RS485 A,output RS485 D,A n c 3 4 D,M16x1 5 D 2. 55 1 n c 1power n c,2 5 shielded,2 1analogue,RS485 B RS485 B. Modbus RTU,M16x1 5 22,D B Modbus RTU,M16x1 5 n c,M16x1 5 RS485 B. D M16x1 5 D Modbus RTU,M16x1 5 n c RS485 A,Modbus RTU D Modbus RTU. M16x1 5 2x M16x1 5,Modbus RTU Modbus RTU power supply. Modbus RTURTU,Modbus analogue output,Option E6 Option E6 RTU. Option E6 Option E6 RS485 B,Option E6 M16x1 5 D,Modbus RTU. GND RS485 GND Output,Modbus RTU shielded,GND RS485 GND RS485. power supply GND Output,Modbus RTU analogue output. power supply Modbus,n c RTU shielded,power supply RS485 A. OUT 2 shielded,GND GNDRS485,Modbus RTU power output. E12analogue,Modbus RTU,Modbus RTU power,analogue output. shielded RS485 A,n c 2 4 3 GND Output,shielded GND A. RS485 OUT 2,power supply,analogue output n c 3 5 4. analogue output,n c 2 34 5 43 GND D,RS485 A OUT 2,Option E12. Option E12,Option E12,E12 5414 RS485,Option E12 D 2 55 1 n c RS485 B V 2155 12 OUT 1. RS485 BModbus 22,supply n c,RS485 PowerV supply analogueOUT. Operating instructions for EE310 Humidity Temperature. shielded 1,Transmitter,D analogue output D OUT 2,Modbus RTU Power supply. n c Modbus analogue,RTU output,RS485 A Power sup,1 x M16x1 5 mounted. Modbus RTU,power supply,power supplyRS485,Modbus RTU. 1 x M16x1 5 enclosed analogue output,M16x1 5 powerRS485. Modbus RTU GND order code J3 GND Output,Modbus RTU analogue. analogue output D,Modbus RTU power supply shielded analogue output. Modbus RTU,Option E6 5,power supply,analogue output RS485 A OUT 2 GND100 240 V AC. Option Option AM3,Polycarbonate,Option E5 E6 enclosure. E12 M16x1 5,Stainless steel enclosure,D Option E6,D Pin assignment 4. Option AM2,E6 Option n c AM3,5 RS485 A n c,5 neutral wire. Modbus RTU 2 3 14 n c,RS485 B Modbus RTU,power supply 4. shielded D,Option E6 RS485 B 2 1,analogue output,phase n c A. RS485 OUT 2,D Modbus RTU,Option E12,Power supply,RS485 B L1. analogue output,n c D phase 14,on for stainless steel housing HS2 D. 5 GND Output L1,Modbus RTU power supply,shielded 2 1 240. Power supply 100,for polycarbonate,Modbus RTU 2xenclosure. Power supply,RS485 GND Output,Option E4 analogue output Option AM3. n c GNDoutput,analogue Output,D OUT 2 GND for polycarbo. RTU power supply shielded,power supply,outputanalogue output. Modbus RTU,Option E12,Modbus RTU,Modbus RTU GND neutral wire. Power suppl,analogue output OUTn c 2analogue RS485 A. 243 134 14 23,E12 power supply output D N,OUT 2 n c OUT 1. Option E12 n c phase,Modbus RTU RS485 B power supply GND Output. GND RS485 L1,Modbus with order code J3 D Modbus RTU phase. power supply n c 1 V OUT 1,RS485 shielded,Bwith order code J3 OUT L1. Option AM3 analogue output n c,only OUT 2 GND,n c RS485 A Power supply analogue output. Modbus powerRTU RS485,D n c A n c,n c OUT 1 wire,Modbus RTU analogue output D. Option E12 Modbus RTU D,Modbus RTU Power supply analogue N output. power supply only with Analogue output Power supply 100 240 V AC. analogue output order code J3,for stainless,Analogue output 1 2. steel enclosure,OUT 1 Power supp,B output 2,power supply. for stainles,power supply D Option AM3 100 240 V AC L1. Modbus RTU analogue output Modbus RTU Powern c supply analogue output RS485 A n c. Option E6 Option AM2 power supply Modbus,Option AM2. Power supplyNO100 240 V AC1 4 D, AM2 Modbus RTU analogue output Option AM2 for polycarbonate enclosure. REL 2 C GNDGND RS485,Output C 2,analogue shielded RS485 B phase C. Option AM3 REL 2 C 2,output GND RS485 D NC neutral wire L1. n c 3 n c 3,power supply shielded RS485 A neutral wire. power supply NC,Modbus RTU AM3analogue output D,Modbus RTU N 3. n c RS485 A phase n c neutral wire,Modbus RTU,NC 4 NO 4 for polycar. only with order code J3,2x M16x1 5,1 2 52 14 n c,power supply REL 1n c B. RS485 2x C GND RS485 5,analogue phase C PE 5,2x M16x1 5. analogue output,Modbus RTU,RS485 B NO 2 1,output shielded. power supply n c,n c RS485 Power neutral,D Analogue Modbus RTU. output Power L1 supply,supply 100,2406VVAC AC,Standard 3. Modbus RTUoperation mode 4 D forInverted,stainless steel enclosure. polycarbonate N,only withModbus,order codeRTU,J3 Power supply. Standard 5,operation 240 V AC Inverted oper,GND Output. polycarbonate 1,output GND Output neutral wire,Option AM2 analogue D OUT 2 n c L1. power supply,phase Analogue output neutral wire Power su. Modbus RTU OUT 2 Modbus RTU n c Power supply 100 240 V AC for stainl. power supply REL 2 C, only with order code J3 C 2 for L1 polycarbonate 2. OUT 1 phase,Modbus RTU n c,GND Output L1 PE,onlyanalogue. with order code J3 NO 3 NC 3,output Option,n c AM2 1 2n c. OUT 1 grounding,Analogue output Power supply NC 240 1. 100 N V AC N,power supply NC,3 4 for NO,stainless steel 4. Modbus2x RTUM16x1 5 REL 2 Power supplyC 100 2402V AC. REL 1 C L1 C,only with order code J3 5 for stainless 5. steel enclosure,Option AM2 6 6,Standard operation mode Inverted. Poweroperation,supply mode,100 240 V4AC,Option AM2 Analogue output NC N. 21 for stainless,NO steel enclosure,2x M16x1 5,Option AM2 NO 3 NC 3. NC 14 Standard NOoperation mode,NO 14 Inverted op,2x M16x1 5. 2x M16x1 5,NOC 365 NC,StandardNO,operation mode6 Inverted operation. 2x M16x1 5 Standard operation mode Inverted operation mode. REL 1 C 5 C 5, Operating instructions for EE310 Humidity Temperature Transmitter 11. power supply,Option E6 RTU,Modbus analogue output,onnection for stainless steel housing HS2 power. GND supply,Modbus RTU analogue output,Option AM3enclosure. Option Option, Polycarbonate Stainless steel enclosure Pin assignment. E4 Option AM3,n c RS485 A n c neutral wire,GND RS485 N. Option AM3,RS485 B 2 1,GND RS485 n c n c,phase 1 n c A. GND Output,Modbus RTU power supply D,analogue output RS485 A OUT 2 GND. Option E12,RS485 for 4,Power supply 100 240,B polycarbonate enclosure 31 phase 1. Modbus RTU,only with 5 D L1,GND Output,power supply. analogue output,order code J3,RS485 B 2 1,Modbus RTU. analogue output neutral wire,D OUT 2 n c for polyc. power supply N,power supply,Modbus RTU,Modbus RTU,100 240 V AC Power supply. phase analogue output,GND Output,Option E6with order code J3. only Option AM2,output 5 L1,OUT 1 OUT 2 n c grounding. power supply,Modbus RTU,only with order code J3 5,Power supply 100 240 V AC. Modbus RTU,power supply,analogue output,Analogue output. stainless steel OUT 2,GND RS485 Analogue output Power s. 2x M16x1 5 for stain,Option AM3 output,analogue NC 1 NO 1. Modbus RTU n c RS485 A n c neutral wire,Option AM2. NO 3 NC NC 1 3,2 1 n c 1 n c,RS485 B phase,2x M16x1 5 NO 3. Modbus RTU,REL 1 C Power supplyC100 240 V AC,5 for polycarbonate enclosure5. NO GND Output 6 NC NC 4 6,analogue 2x M16x1 5,output InvertedCoperation mode. OUT 2Standard operation,REL 1 neutral,power supply. Modbus RTU,3 phase NO 6,only with order code J3 5 L1. Standard operation mode Inverted o,n c 1 2 OUT 1 grounding. Analogue output Power supply 100 240 V AC,for stainless steel enclosure. External diameter of the supply cable for option AM3 10 12 mm 0 39 0 47. Option AM2, Maximum wire cross section for AM3 connecting cable 1 5 mm AWG 16. External diameter of the cable,2 Modbus RTUC and analogue 2. output female plug,C 4 6 mm 0 162 0 24, Maximal wire cross section for connecting cable 0 5 mm AWG 21. 2x M16x1 5,REL 1 C 5 C 5,Standard operation mode Inverted operation mode. 12 Operating instructions for EE310 Humidity Temperature Transmitter. 4 3 Probe mounting wall duct version,Fig 7 ounting of model EE310 T1 The probe. M Fig 8 ounting of model EE310 T2 The probe, shall point downwards shall be mounted horizontally or vertically. pointing downwards,4 4 Mounting remote sensing probe of EE310 T5. For mounting the probe into a separation wall use the stainless steel mounting flange The immersion. depth is adjustable The probe shall be mounted horizontally. drip water,protection,temperature gradient along bow for draining 85. the probe shall be avoided condense water 3 3,probe into separation wall hanging probe. Fig 9 Mounting the remote probe of EE310 T5, For accurate measurement it is of paramount importance to avoid T gradients along the probe In case. of large T difference between the front and the back of the probe it is highly recommended to insert the. probe completely into the process Should this not be possible place a thermal isolation layer on the. part of the probe outside the process on the cable side. The stainless steel mounting flange is not appropriate for pressure tight mounting For pressure tight. requirements use EE310 T10, For probe hanging onto its cable from the ceiling in applications where condensation is likely to happen. it is important to avoid condense water getting from the cable to the probe and into the sensing head. For this use the drip water protection see chapter 8. Operating instructions for EE310 Humidity Temperature Transmitter 13. 4 5 Mounting pressure tight probe of EE310 T10,General safety instructions for installation. The installation commissioning and operation of the EE310 T10 may be performed by qualified staff. only Special attention shall be paid to the correct installation of the probe into the process In case of. inappropriate installation there is the risk for the probe to be suddenly expulsed due to the pressure in. the process, Bending over the sensing probe should be avoided under any circumstances. Make sure that the surface of the probe is not damaged during installation A damaged probe surface. may lead to damaged seals and consequently to leakage and pressure loss. The probe is rated with leakage rate B according to EN12266 1. 4 5 1 Installation of the probe directly in the process see Fig 10. For direct probe installation shut off valves shall be placed on both sides of the probe insert This allows. the sensor probe to be easily removed for maintenance and calibration. For direct installation into a pressure chamber make sure that the pressure in the chamber and the. ambient pressure are equal before removing the probe The temperature during probe installation may. deviate by max 40 C 72 F from the regular temperature during normal operation. Replace the metal sealing ring see Fig 10 by a new one every time before re installing the probe. Probe installation steps,Close both shut off valves. Place the sensor probe into the probe insert and adjust the immersion depth. Tighten the lock nut with a torque of 30 Nm,Open the shut off valves. Observe strictly the tightening torque A torque lower than 30 Nm results in a smaller retention force of. the clamping sleeve This leads the risk of sudden expulsion of the sensing probe due to the pressure. A torque higher than 30 Nm may lead to permanent deformation of the clamping sleeve and the sensing. probe This would make the removal and re installation of the probe difficult or even impossible. fixed installation pressure tight up,to 20 bar 300 psi. metal sealing ring,1 2 ISO or 1 2 NPT,inside diameter. 13 mm 0 55,stop valve stop valve, Fig 10 Installation of the EE310 T10 probe directly into the process. 4 5 2 Installation of the probe with ball valve set. The ball valve set allows for installation and removal of the probe without process interruption. For mounting into a duct the ball valve shall be installed perpendicular to the flow direction. The two metal sealing rings see Fig 11 shall be replaced every time prior to re installing the probe. The temperature during probe installation may deviate by max 40 C 72 F from the regular tempera. ture during normal operation, 14 Operating instructions for EE310 Humidity Temperature Transmitter. Installation of the probe see Fig 11, Install the probe into the ball valve while the ball valve is closed. Open the ball valve, Slide the probe through the ball valve to the desired immersion depth Depending on the process. pressure additional tools may be necessary for pushing the probe into the process Make sure not to. damage the probe and the cable,Tighten the lock nut with a torque of 30 Nm. Observe strictly the tightening torque A torque lower than 30 Nm results a smaller retention force of the. clamping sleeve This leads the risk of sudden expulsion of the sensing probe due to the pressure A. torque higher than 30 Nm may lead to permanent deformation of the clamping sleeve and the sensing. probe This would make the removal and re installation of the probe difficult or even impossible. metal sealing ring,as a standard contained,with the ball valve set. metal sealing ring,as a standard contained,with the probe. 1 2 ball valve,is not available for clamping sleeve. an NPT thread,sealing element 2,sealing element 1,adapter body. inside diameter,13 1 mm 0 55, Fig 11 Installation of the probe by utilizing the ball valve set. Removing the probe, Hold firmly the probe to avoid it being suddenly expulsed when releasing the lock Do not bend dam. age the probe cable, Loosen slowly the lock nut with a spanner spanner width 24 only till the probe is pushed out by the. overpressure in the process Do not completely loosen the lock nut but only as much as necessary. for the probe to slide, After the probe has been pushed out of the process up to the stop close the ball valve. Remove the probe from the ball valve, Observe the correct positioning of the sealing element 1 before reinstalling the probe. Operating instructions for EE310 Humidity Temperature Transmitter 15.
21-Feb-2020 5 Views 14 Pages
EA 9777-2 Page 2 ELECTRONIC ASSEMBLY reserves the right to change specifications without prior notice. Printing and typographical errors reserved. Documentation of ...
21-Feb-2020 6 Views 135 Pages
P4200.11 Facilities Operations Manual (4/12/2016) This Program Statement incorporates changes in the areas listed below: Rewrite of Chapter 1, Facilities Administration and Organization Rewrite of Chapter 2, Work Programming, Scheduling and Reporting Rewrite of Chapter 3, Buildings and Facilities Projects
21-Feb-2020 5 Views 83 Pages
Guidelines on the Calibration of Non-Automatic Weighing Instruments EURAMET/cg-18/v.02 January 2009 Previously EA-10/18 European Association of National Metrology ...
21-Feb-2020 5 Views 53 Pages
was shared with stakeholders whose input was gathered during the planning process as part of a desire to generate national ownership and support for the plan. Over 200 stakeholders from parents and teachers to students and members of the general public were involved in the planning process.
21-Feb-2020 5 Views 40 Pages
alanwire.com 830 South West Street / Sikeston, MO 63801 8003258050 8006223355 ABOUT US Alan Wire Company began in 1974 as a small bare copper manufacturer serving the
21-Feb-2020 5 Views 11 Pages
Penyaluran tenaga listrik bawah tanah yang sekarang ini banyak dibangun harus didukung dengan sistem isolasi ... penyambungan kabel dimungkinkan adanya
21-Feb-2020 5 Views 11 Pages
Aluminium merupakan konduktor listrik yang ... atau dengan teknik penyambungan ... maka aluminium sangat baik untuk kabel-kabel listrik overhead maupun bawah tanah.
21-Feb-2020 5 Views 8 Pages
CBSE 12th Physics 2017 Unsolved Paper Delhi Board Buy solution: ... Q.12. Write the two processes that take place in the formation of a p-n junction. Explain with the help of a diagram, the formation of depletion region and barrier potential in a p-n junction. 3 marks Q.13. (i) Obtain the expression for the cyclotron frequency. 3 marks (ii) A deuteron and a proton are accelerated by the ...
21-Feb-2020 6 Views 6 Pages
CBSE 12th Physics 2016 Unsolved Paper Delhi Board TIME - 3HR. | QUESTIONS - 26 THE MARKS ARE MENTIONED ON EACH QUESTION _____ SECTION-A Q.1. A point charge +Q is placed at point O as shown in the figure. Is the potential
21-Feb-2020 5 Views 63 Pages
Dasar-Dasar Pemeliharaan Instalasi Listrik 8 NYM-O Kabel thermo plastik berperisai logam ... saluran kabel dan di bawah tanah untuk instalasi ind ustri dan ...